The sensors of the PCB-261 series are ready-to-use force measuring elements with measuring ranges between 0.045 and 222 kN. At the same time they measure the dynamic and quasistatic forces in three orthogonal axes. The sensors are already preloaded and therefore quick and easy to assemble. They are available with integrated ICP® preamp or charge...
The sensors of the PCB-261 series are ready-to-use force measuring elements with measuring ranges between 0.045 and 222 kN. At the same time they measure the dynamic and quasistatic forces in three orthogonal axes. The sensors are already preloaded and therefore quick and easy to assemble. They are available with integrated ICP® preamp or charge output.
Triaxial ICP® power measurement element Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 2.22 kN, Sensitivity (Z-axis) 0.56 mV / N, Sensitivity (X - and Y-axes) 2.25 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side Triaxial ICP® power measurement element Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 2.22 kN, Sensitivity (Z-axis) 0.56 mV / N, Sensitivity (X - and Y-axes) 2.25 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side
Triaxial ICP® power measurement element Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 4.45 kN, Sensitivity (Z-axis) 0.56 mV / N, Sensitivity (X - and Y-axes) 1.12 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side Triaxial ICP® power measurement element Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 4.45 kN, Sensitivity (Z-axis) 0.56 mV / N, Sensitivity (X - and Y-axes) 1.12 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side
Triaxial ICP® power measurement element Measuring range (Z axis) 44.5 kN, Measuring range (X - and Y-axes) 17.8 kN, Sensitivity (Z-axis) 0.06 mV / N, Sensitivity (X - and Y-axes) 0.28 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side Triaxial ICP® power measurement element Measuring range (Z axis) 44.5 kN, Measuring range (X - and Y-axes) 17.8 kN, Sensitivity (Z-axis) 0.06 mV / N, Sensitivity (X - and Y-axes) 0.28 mV / N, Frequency range (Z axis) 0.01 … 10,000 Hz, Frequency range (X - and Y-axes) 0.001 … 10,000 Hz, 4-pin female connector on the side
Triaxial strength measuring element with charge output, Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 2.22 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side Triaxial strength measuring element with charge output, Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 2.22 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side
Triaxial strength measuring element with charge output, Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 4.45 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side Triaxial strength measuring element with charge output, Measuring range (Z axis) 4.45 kN, Measuring range (X - and Y-axes) 4.45 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side
Triaxial strength measuring element with charge output, Measuring range (Z axis) 44.5 kN, Measuring range (X - and Y-axes) 17.7 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side Triaxial strength measuring element with charge output, Measuring range (Z axis) 44.5 kN, Measuring range (X - and Y-axes) 17.7 kN, Sensitivity (Z-axis) 3.4 pC/N,. Sensitivity (X - and Y-axes) 7.2 pC/N,. Upper frequency limit 10,000 Hz,. Three 10-32 connectors on the side